TitleBiomechanical properties of the thin PVD coatings defined by red blood cells
Journal titleBulletin of the Polish Academy of Sciences: Technical Sciences
Divisions of PASNauki Techniczne
Date2015[2015.01.01 AD - 2015.12.31 AD]
ReferencesTakaseya (2007), Aortic valve replacement with mm St Jude Medical prostheses for a small aortic root in elderly patients, Thorac Surg, 2050, doi.org/10.1016/j.athoracsur.2007.01.044 ; Dadsetan (2001), In vitro studies of platelet adhesion on laser - treated polyethylene terephtalate surface, Mater Res, 540. ; Borger (2006), Twenty - year results of the Hancock II bioprosthesis, J Heart Valve Dis, 16, 49. ; Varon (1998), Cone and plate ( let ) analyzer : monitoring glycoprotein IIb / IIIa antagonists and von Willebr and disease replacement therapy by testing platelet deposition under flow conditions, Heart J, 8, 135. ; Borger (2007), Stentless aortic valve reoperations : a surgical challenge, Thorac Surg, 20, 737, doi.org/10.1016/j.athoracsur.2007.04.061 ; Aagaard (2005), and Fifteen years clinical experience with the CarboMedics prosthetic heart valve, Heart Valve Dis, 14, 82. ; Stefanidis (2005), Cannie re - year experience with the ATS mechanical valve in the mitral position, Thorac Surg, 27, 1934, doi.org/10.1016/j.athoracsur.2005.01.002 ; Emery (2005), The St Jude Medical cardiac valve prosthesis : a - year experience with single valve replacement, Thorac Surg, 23, 25. ; Bach (2005), Ten - year outcome after aortic valve replacement with the freestyle stentless bioprosthesis, Thorac Surg, 22, 480, doi.org/10.1016/j.athoracsur.2005.03.027 ; Jones (1996), Methods for in vitro analysis of leukocyte adhesion under flow condition inWeir s Handbook of Experimental Immunology, Science, 9. ; Riess (2007), Hemodynamic performance of the medtronic mosaic porcine bioprosthesis up to ten years, Thorac Surg, 18, 1310, doi.org/10.1016/j.athoracsur.2006.07.034 ; Bottio (2004), Small aortic annulus : the hydrodynamic performances of five commercially available bileaflet mechanical valves, Cardiovasc Surg, 28, 457. ; Otto (1997), Dymanic blood cell contact with biomaterials : validation of a flow chamber system according to international standards, Mater Sci Mater Med, 8, 119, doi.org/10.1023/A:1018515001850 ; Seyfert (2002), In vitro hemocompatibility testing of biomaterials according to the ISO, Eng, 19, 10993. ; Yago (2006), Catch bonds govern adhesion through L - selectin at threshold shear, Cell Biology, 10, 166. ; Cohen (2002), Are stentless valves hemodynamically superior to stented valves ? A prospective randomized trial, Thorac Surg, 19, 767, doi.org/10.1016/S0003-4975(01)03338-0 ; Demilly (2006), Kinetics of yeast detechment from controlled stainless steel surfaces, Colloids Surf, 51. ; Huang (2003), Hemocompatibility of titanium oxide films, Biomaterials, 24, 2177, doi.org/10.1016/S0142-9612(03)00046-2 ; Mueller (2012), Dynamic in vitro hemocompatibility testing improving the signal tonoise ratio, Tech, 1. ; Lackner (2010), pulsed laser deposited coatings on polymers, Biomedizinische Technik, 55, 57, doi.org/10.1515/bmt.2010.001 ; Tominaga (2005), - year experience with the Carbomedics cardiac prosthesis, Ann Thorac Surg, 24, 10. ; Valfrè (2006), Clinical results of Hancock II versus Hancock Standard at long - term follow - up, Cardiovasc Surg, 15, 132. ; Totaro (2005), Edwards PERIMOUNT Magna bioprosthesis : a stented valve with stentless performance ?, Cardiovasc Surg, 1668. ; Lopez (2008), Does the use of stentless aortic valves in a subcoronary position prevent patient - prosthesis - mismatch for small aortic annulus, Surg, 21, 331.